08 Aug How to Titrate Your Narval Device
How to Titrate Your Narval Device
To titrate your Narval Device, you first need to remove your current titration. To do so, adjust the bars of the titration so that they are in line with the hole in the bottom mouth piece, and proceed to pop it out. You will repeat this process of aligning the titration and popping it out of the mouth piece hole for the top half. Once the old titration is completely removed, you will begin adding the new titration by popping it into the top mouth piece. Next, you will connect the top piece to the bottom piece by putting the titration through the bottom half’s hole. Upon locking the titration into place, close the mouth piece until the top and bottom rows are touching. The titration should automatically rotate to an unaligned position to the holes, and the Narval Device will be in a rested state. This final step will ensure that you have properly replaced the titration in your Narval Device.